Pulsoksimetr KMP-Ro60 can be used for measurement of a pulse saturation of oxygen and pulse rate on a finger. The product is suitable for use in house and hospital conditions, oxygen bar, when rendering the medical and sanitary help to the population and medical examination in sport (It can be used to or after sports. It is not recommended to use the device during sports), etc. The pulse saturation of oxygen (saturation by oxygen) represents percentage of oxyhemoglobin (HbO2) in a total amount of hemoglobin (Hb) in blood, so-called content of oxygen (O2) in blood. It is the important biological parameter of breath. Therefore, exact information on a saturation of patients gives significant assistance to doctors in detection of potential danger and is of great importance in the field of clinical medicine. Pulsoksimetr has the compact size, low electricity consumption, is convenient in use and portable. The patient needs to pass only one finger throughout the sensor for diagnostics, and on the display screen at once will appear value of pulse saturation by oxygen with high precision and high frequency of repetition.
- Simple and convenient management
- The product has the compact size, is easy and convenient at transfer.
- Low electricity consumption.
- Display of indicators SpO2.
- Display of indicators of pulse rate, histogram display.
- Pulse form display.
- Indicator of low power: The indicator of low power appears before the device begins to work incorrectly owing to low power
- Function of automatic shutdown: The product is automatically disconnected in the absence of a finger in the device more than 5 seconds when the device is in a condition of the measuring interface.
|Display format||Digital screen display|
|Measurement of SpO2||Range: 0% ~ 100% Accuracy: ±2% in the range of 70 ~ 100% of SpO2; value below of 70% is not defined.|
|Pulse rate measurement||Range: The 30th blow in min. ~ the 250th blow in min.
Accuracy: ±2 blows in mines or ±2% (big size)
|Clearness||1% for SpO2 and 1 beat per minute for pulse rate|
|Working tension||3.6 - 4.2 In, the recharged lithium battery × 1|
|Light-sensitive sensor||Red light (wavelength – 660 nanometers ~ 6.65 MW)
Infrared beam (wavelength – 880 nanometers, 6.75 MW)
|Range of a sound alarm signal:||SpO2: 0% ~ 100% Pulse rate: 0 blow in min. ~ the 254th blow in min.|
|Pulse saturation by SpO2 oxygen||Two-place digital organic LED display|
|Pulse rate of PR||Three-digit digital organic LED display|
|Intensity of pulse (histogram)||Gistogrammny organic LED display|
|Average value||To calculate average value on everyone to 4 measured values. The deviation between average value and the actual value does not exceed 1%.|
|Average pulse rate||To calculate average pulse rate on each 4 cycles of heartbeat. The deviation between average value and the actual value does not exceed 1%.|
|Dimensions||57 (D) × 32 (Sh) × 30 (IN)|
|Weight||about 50 g (together with lithium batareykoy*1) mm|